From HacDC Wiki
HacDC is in the process of refurbishing a scanning electron microscope: a Cambridge Stereoscan 200.
March 31st: Cleaned the plinth and took photographs of the vacuum components. Available here. -Dan K.
Oil diffusionTurbo molecular pump was hanging due to the lip falling out of the rubber suspension. Could be cleaned and probably popped back in.
- Broken switch on roughing pump was bypassed internally. Plugging in the outlet should activate the pump. (This pump normally runs 24/7 when the machine is operational.)
- Compressed air needed to get more dust out
March 30th: Moving complete! The SEM is now in the workshop
March 29th: Awaiting pick-up and installation (March 29th and 30th).
March 23rd: A space in the workshop has been cleared. Photo: http://i.imgur.com/BDNWL5z.jpg
Maintenance and repair logs will be kept on an ongoing basis. For previous status and repairs, see:
- The original S200 operating instructions will be available at the space in a three-ring binder near the machine.
- If possible, members will also have access to a PDF copy of Scanning Electron Microscopy - A Student's Handbook, 1980, self-published by Michael Postek, et al. This book was widely used by microscopists and is still regarded as a good learning reference, but the original binding is rare and prone to falling apart. It might be possible to use the PDF copy for private, educational use.
- SPI supplies: http://www.2spi.com
- O-ring supplier: http://www.parker.com
We'll have a list here of HacDC members who are responsible for the SEM, including people that should be notified about problems.
- Apiezon M vacuum grease for "O" rings, to be used for vacuum seals at room temperature: http://www.2spi.com/catalog/vac/apiezon/apiezon-m.php ($70 for 25g). Edit: page 058 of the manual states "The use of O ring grease is not recommended on any O rings, but a small amount may be used, if necessary, on moving seals e.g. the specimen stage and apereture changer micrometers."
- 31Mar2014 - there's a syringe of Ap. T grease in the space (~2g). It's on top of the SEM's chamber atm.
- Replacement "O" ring for chamber->gun vacuum tube, for the chamber-side mounting adapter. (Will this work? http://www.grainger.com/product/O-Ring-1RHY8)
- Inland 19 pump oil for Edwards E2M-12 roughing pump: http://www.sisweb.com/vacuum/sis/inland.htm ($30 per gallon) (needs confirmation that this oil is suitable)